Panametrics' Customer Story
Major multinational food and drink processing company requested support to validate
the performance of existing electromagnetic flow meters, that were failure-prone, in
their water treatment plants. The liquid clamp-on flow meters were used to verify the
plant balance for the inline meters out of accuracy specifications or out of order.
Water treatment plant 1” – 6” SS316 lines, mineral water. Temp. 20°C – 50°C
• Hygienic Flow meters
• No Shut down required
• Same meter suitable for all line sizes
• No periodic calibration required
The electromagnetic flow meters at the water treatment plant experienced failures
& performance issues. After replacing several meters the failures continued.
From the discussion with plant process team it appeared that the reliability issues with
the inline flow meters started after they changed the Clean-in-Place process from
chemicals to ozone. Ozone led to electrode oxidization from the mag meters over time.
Portable clamp on flowmeters were used to check some of the meters. The tests
were quite successful.
AquaTrans AT600 meters were installed for two weeks with existing electromagnetic
flow meters to validate the first findings. The AquaTrans AT600 reading was very stable
and within the expected accuracy as per the pump curves according to the customer.
The customer decided to purchase six meters to replace faulty electromagnetic
flow meters and recommended the same solution to be implemented in other
plants within the country.
Learn more about our clamp on flow meters
Follow us on LinkedIn for more information
Download the Application Note here
Read one of our other success stories